EZ1 Test Card
For easy setup of the protocol to evaluate pipetting accuracy of the BioRobot EZ1
The EZ1 Test Card contains protocols designed to evaluate the pipetting accuracy of the BioRobot EZ1. The BioRobot EZ1 pipets either 100 µl or 500 µl of water from one set of tubes to another. Pipetting accuracy should be within defined specifications of ±8 µl for 100 µl, corresponding to pipetted volumes in the range of 92–108 µl and ±40 µl for 500 µl, corresponding to pipetted volumes in the range of 460–540 µl. Another protocol describes how to test the tip adapters for liquid leaks.
EZ1 Cards are used in combination with EZ1 Kits on the BioRobot EZ1. Please refer to the specific EZ1 Kit handbook and the BioRobot EZ1 User Manual for information about product use.
fragment fix placeholder